Directly measured mucosal pressures produced by the i-gel and laryngeal mask airway Supreme in paralysed anaesthetised patients
Eschertzhuber S et al. – The findings suggest that the LMA Supreme mucosal pressures were higher in the hypopharynx than in the distal oropharynx and base of the tongue. There were no pressure differences between the locations for the i–gel.
The i-gel and LMA Supreme are extraglottic airway devices with non-inflatable and inflatable cuffs, respectively.
The authors hypothesised that directly measured mucosal pressures would differ between these devices in anesthetised paralysed patients.
Thirty patients were randomly allocated to receive one of these two devices.
Four pressure sensors were attached to all airway devices used to measure mucosal pressure at the base of the tongue, the distal oropharynx, the hypopharynx and the pyriform fossa.
At these four places, median (IQR [range]) i-gel mucosal pressures were 8.0 (2.7-10.7 [0-26.7]), 5.0 (2.7-7 [1.0-37.3]), 9.3 (2.7-13.3 [0-22.7] and 8.0 (2.7-10.7 [0-25.3])cmH2O, respectively, and for the LMA Supreme, these were 5.0 (0.5-8.0 [0-33]), 4.0 (1.3-9.3 [0-24]), 10.7 (4-17.3 [0-26.7]) and 8.0 (0-10.7 [0-36])cmH2O, respectively.
Mucosal pressures were low and similar for both devices.
MDLinx connects healthcare professionals and patients to tomorrow's important medical news, while providing the pharmaceutical and healthcare industries with highly targeted interactive marketing, education, content, and medical research solutions.